. During the manufacturing process, the instrument errors are carefully determined and set to zero. Insert the compartment cover into the compartment port P3. Description Value Maximum average radiant power 0. Every vision on every scale becomes reality with ease. Keep them for some time inside your jacket or in the vehicle to allow them to adjust to the ambient temperature.
No more runs can be added later. Check the target by looking trough the telescope. Aiming at prisms is only permitted when looking through the telescope. If an external power supply is connected and the internal battery is inserted, then the external power is used. After this storage period the batteries must be recharged. With a complete package of easy-to-use software your work can start immediately and be finished faster than ever before. Shipping When transporting the product by rail, air or sea, always use the complete original Leica Geosystems packaging, transport container and cardboard box, or its equivalent, to protect against shock and vibration.
Check Yes, this time only. Select Add another calibration loop if more runs have to be added. Field adjustment After transport inspect the field adjustment parameters given in this user manual before using the product. Accurate measurements to prisms should be made in Prism mode. The risk of injury for laser class 3R products is limited because of: a unintentional exposure would rarely reflect worst case conditions of e. Precautions: When measuring in remote control mode, always check your results for plausibility.
Starts the edit mode for editable fields. Precautions: All users must follow the safety directions given by the manufacturer and the directions of the person responsible for the product. Remove batteries from the product and the charger before storing. Read carefully through the User Manual before you switch on the product. No more runs can be added later.
A separation distance of at least 20 centimetres should be kept between the antenna and the body of the user or nearby person within the intended application. A separation distance of at least 20 centimetres should be kept between the antenna and the body of the user or nearby person within the intended application. Precautions: While the product is in use, connecting cables, for example product to external battery, product to computer, must be connected at both ends. The standard deviation can be calculated from the second run onwards. If not already installed, run the ActiveSync instal- lation program. Data can be transferred in various ways. Operating times will be shorter when working in cold weather.
Adjustment Please enlarge this target plate to 200% prior to using it for adjustment. Tighten the horizontal adjustment screw d as much as to move the laser beam slightly right of the upper crosshairs on the target plate. PowerSearch PowerSearch refers to the instrument sensor which enables the automatic rapid finding of a prism. For high precision measurements, the relative humidity must be measured and entered along with the air pressure and the temperature. Low operating temperatures reduce the capacity that can be drawn; high operating temperatures reduce the service life of the battery.
Operation by touch screen Information is selected and entered on the screen using the supplied stylus. Your benefit - optimized productivity with exact photo documentation of site conditions to avoid paper notes and costly re-visits. They indicate the basic device status. If an adjustment is necessary due to external influences, return the instrument to any Leica Geosystems authorised service workshop. This person is also accountable for the training and the deployment of personnel who use the product and for the safety of the equipment in use. Non-motorised instruments guide to the other face. Check Install the software automatically Recommended.
Differential phase Static Kinematic in post-processing Horizontal Vertical Horizontal Vertical 5 mm + 0. The procedure can be started in any telescope face. Some other instrument errors and mechanical parts can be adjusted mechanically. Description Value Maximum average radiant power 0. The workflow is the same for keyboard and touch screen entry, the only difference lies in the way information is selected and entered.